So I’m making good progress building out the computer control and data acquisition system for the reactor. Currently we have a manually controlled needle valve to regulate flow on the deuterium handling system. I want to replace this with a computer controlled mass flow controller.
A new mass flow controller costs roughly $1,100. There are plenty available on ebay starting from $70.
A mass flow controller must be calibrated for a specific gas. Some can be calibrated on demand with a digital interface, and some are hard wired to work with a specific gas. I’m wondering if we can get away with using a hydrogen calibration for our deuterium system?
The next parameter is flow rate. Here is some background on flow rate nomenclature: flow rate nomenclature. Although mass flow controllers meter based on mass flow rate, they confusingly are rated using volumetric flow rate metrics like SLPM (standard liters per minute), SCCM (standard cubic centimeters per minute) or SCFH (standard cubic feet per hour). Conversion between the two must be done at standard temperature and pressure, and must take into consideration the density of the gas: details.
So before I can spec a mass flow controller I need to calculate our midpoint flow rate needs.
The chamber holds ~6,500 cm^3
The turbo pump removes gas at the following rates:
N2 -> 56 liters/sec
He -> 48 liters/sec
H2 -> 36 liters/sec
So the mass flow controller meters how much fuel is flowing into the chamber. We also want to control how much fuel the vacuum pump is pulling out of the chamber. This requires a gate valve or butterfly valve. Many of them include an integrated bellows connection (which is great), and are pneumatically actuated (not so great).
Now my naive attempt to calculate flow rate needs:
Without a gate valve the pumps will remove 2160 l/min. Now the typical mass flow controller seems to max out at 10 SLM (l/min)…. so the pump will remove fuel far faster than the mass flow controller can supply it. And this will waste expensive fuel.
I think these calculation work for pressures in the laminar flow region (> 1e^3 torr). We are working with pressures between the laminar flow domain and the free molecular flow domain – I don’t know how the calculations change at this point.
ps. Wolfram Alpha seems to be useful for density calculations.